Finger Pulse Oximeter Spo2/PR Fingertip Oxygen Monitor SPO2 PR Home Health Care *MEDCERT*
– Color OLED display, simultaneous display for testing value and plethysmogram. – The display interface of OLED can rotate four directions with six different display modes after pressing the power button for less than 0.5s. – Real-time spot-checks – Pulse waveform & barograph display – Advanced setting for alarm – Functions of audio alarm and pulse sound – Automatic power off – Low power consumption down to 0.3% – 50 hours continuous to work – Low perfusion ≤0.4% – Low voltage indicator – Widely used in hospital, home healthcare, oxygen bar, community medical centre, alpine area, sports healthcare etc.
1.5V (AAA size) Not included alkaline battery x 2 Supply voltage: 2.6~3.6V
Automatically power off when no signal in the oximeter for more than 8 seconds
Dimension & Weight
58 (L) × 31(W) × 29(H) mm 60g
1 x RPO-8B3 Fingertip Oximeter
1 x Lanyard
1 x English user’s manual
How to use it:
Turn on the pulse oximeter by firmly pressing the power button. The screen should light up almost instantly.
Place the sensor — the part that opens and closes like a clothespin — on any finger, with the sensor screen above the fingernail. If the sensor doesn’t have a screen, run the cable along the back of the finger or hand. Don’t use the thumb because readings are less reliable than finger readings.
Wait quietly while the pulse oximeter acquires a signal. This may take 10 seconds or more, depending on the device and the conditions. Excessive movement during measurement can decrease the accuracy of the result or may cause an error message.
Look at the display to see the heart rate, usually indicated with a heart or pulsing light. The percent of oxygen saturation is typically indicated by the symbol “SpO2.” Many devices also have a pulse tone that beeps in time with the heart rate.
Leave the sensor on for continuous monitoring. The sensor can become uncomfortable or cause pressure sores if left on the finger too long. Check and/or move the sensor at least every two to four hours. If only a single measurement is required, remove the sensor and press the power button to turn off the device.
To ensure proper function, read the manual for your specific pulse oximeter.
Remove nail polish, especially dark colors, before placing the sensor on the finger.
Place the sensor on the finger before turning on the pulse oximeter for a faster result.
Pulse oximeters are less accurate if there is poor blood flow to the finger. This may occur if the person is cold, has very low blood pressure or is having a heart attack.
Carbon monoxide — due to smoke inhalation, carbon monoxide poisoning or heavy cigarette smoking — results in falsely high pulse oximeter readings.
Altitude may result in lower percent oxygen saturation due to the lower oxygen pressure in the air, particularly if you have recently arrived.
A normal percent oxygen saturation reading is in the 95 to 100 percent range. In the event of a low oxygen saturation measurement, look for signs of respiratory distress. This includes shortness of breath, wheezing, difficulty breathing or a bluish discoloration of the face, lips or fingernails.
Seek medical attention right away if the respiratory distress is a new or uncomfortable symptom. if it’s getting worse or if the pulse oximeter measurement is below 90 percent.